Electromagnetic flowmeter is only applicable to measure the flow of conductive liquid, which is wide used in water supply, sewage water measuring, industry chemical measuring etc. The remote type is with high IP protection class and can be installed in different locations for the transmitter and converter. Output signal can pulse,4-20mA or with RS485 communication.
Magnetic flow meter works based on Faraday’s law, and measure conductive medium with conductivity more than 5 μs/cm and flow range from 0.2 to 15 m/s. An Electromagnetic Flowmeter is a volumetric Flowmeter that measures the flow velocity of a liquid through a pipe.
The measurement principle of magnetic flowmeters can be described as follows: when the liquid goes through the pipe at the flow rate of v with a diameter D, within which a magnetic flux density of B is created by an exciting coil, the following electromotive E is generated in proportion to flow speed v:
Noted: the product strictly prohibited to be used in explosion-proof occasions.
Domestic water, industrial water, raw water, ground water, urban sewage, industrial wastewater, the processed neutral pulp, pulp slurry, etc
|Electrode material||Stainless steel SUS316, Hastelloy C, Titanium,|
|Medium temperature||Integral type: -10℃~80℃|
|Split type: -25℃~180℃|
|Power supply||100-240VAC，50/60Hz, 22VDC—26VDC|
|Electrical conductivity||Water 20μS/cm other medium 5μS/cm|
|Structure type||Tegral type, split type|
|Product standard||JB/T 9248-1999 Electormagnetic Flowmeter|